From Vancouver, Washington, USA:
We are having a problem deciding which blood glucose machine to use. We started out with the One Touch Profile and then we bought the Glucometer Elite. I love the Glucometer Elite because our son is 2 1/2 years old. Yesterday I tested my son on both machines and the Elite was 80 and the Profile was 162. I then did several tests and they were closer (98/109). How do I know which machine is right? I ran tests on both and they were working correctly. How do I find out which machine is the best one?
All the meters are accurate when used properly. Be sure you've not getting air bubbles into the Elite strips; this can alter the readings. You can try using "Barbie basters" (our program's nickname for the little capillary tubes available from Lifescan) to collect samples for the Profile. At two-and-a-half years old, a lot of kids can use the Profile themselves with a little help and encouragement; the Elite might be harder for a youngster to use.
P.S. there's really no difference between values of 98 and 109.
[Editor's comment: The things one learns! I never thought that the term "Barbie baster" would come into widespread usage, but Steph informs me that it's used by other pediatric diabetes programs, also. I always thought the clear plastic little squeeze thing looked like the float for Barbie's toilet tank, but the concept of a baster makes a bit more sense. Oh well. WWQ]
Original posting 21 Feb 97
|Return to the Top of This Page|
Last Updated: Tuesday April 06, 2010 15:08:54
This Internet site provides information of a general nature and is designed for educational purposes only. If you have any concerns about your own health or the health of your child, you should always consult with a physician or other health care professional.
This site is published by T-1 Today, Inc. (d/b/a Children with Diabetes), a 501c3 not-for-profit organization, which is responsible for its contents. Our mission is to provide education and support to families living with type 1 diabetes.
© Children with Diabetes, Inc. 1995-2018. Comments and Feedback.