From Medford, Massachusetts, USA:
I'm trying to find out are specifics on the chemistry used in different blood glucose meters. As more companies shift to electrochemical technologies I'm wondering how the technologies differ and even if they differ. Are they all based on oxidised glucose or peroxide or do they use other methods? Which do what and which are better?
Meters on the market to read blood glucose usually do so by using a chemical reaction. The amount of glucose in the blood sample is then "measured" through an enzyme reaction with glucose oxidase. This either becomes a color that is read by the meter system or, most recently, has the meter respond to how much of a minute electric current is produced -- then the meter translates that current to an equivalent blood glucose reading. All of these systems are proprietary. You may want to visit each of these company websites and see if there is more detailed information available.
If you go to PubMed, you could also look at the published systems in the medical literature, or, if you have access, in more detailed electrochemical engineering and clinical chemistry journals.
Original posting 23 Jan 2001
Posted to Blood Tests and Insulin Injections
Last Updated: Tuesday April 06, 2010 15:09:18
This Internet site provides information of a general nature and is designed for educational purposes only. If you have any concerns about your own health or the health of your child, you should always consult with a physician or other health care professional.
This site is published by T-1 Today, Inc. (d/b/a Children with Diabetes), a 501c3 not-for-profit organization, which is responsible for its contents. Our mission is to provide education and support to families living with type 1 diabetes.
© Children with Diabetes, Inc. 1995-2018. Comments and Feedback.